Note:Could be customized under your request
1. High sensitivity, simple, easy.
2. Accuracy: >99%.
3. CE & ISO approved.
4. Specimen: Urine.
5. Model No:MTD-U02D
The MTD One Step Methadone Test Cassette is a lateral flow chromatographic immunoassay for the detection
of Methadone in urine at a cut-off concentration of 300 ng/mL. This test will detect other opiates. Please
refer to analytical specificity table in this package insert.
Methadone is a narcotic pain reliever for medium to severe pain. It is also used in the treatment of heroin
(opiate dependence: Vicodin, Percocet, Morphine, etc.) addiction. Oral Methadone is very different than IV
Methadone.Oral Methadone is partially stored in the liver for later use. IV Methadone acts more like heroin.
In most states you must go to a pain clinic or a Methadone maintenance clinic to be prescribed Methadone.
Methadone is a long acting pain reliever producing effects that last from twelve to forty-eight hours. Ideally,
Methadone frees the client from the pressures of obtaining illegal heroin, from the dangers of injection, and
from the emotional roller coaster that most opiates produce. Methadone, if taken for long periods and
at large doses,can lead to a very long withdrawal period.The withdrawals from Methadone are more
prolonged and troublesome than those provoked by heroin cessation, yet the substitution and phased
removal of methadone is an acceptable method of detoxification for patients and therapists.1
The MTD One Step Methadone Test Strip yields a positive result when the Methadone in urine exceeds
· Specimen: Urine
· Cut-off:300ng/ml or 200ng/ml
· Usage: home-testing, self-testing, professional-testing
· Inner Packing: 1 test/pouch with desiccant, cassette with dropper
· Certificate: CE, ISO & FDA
· Accurate Results in 5 Minutes
· Convenient Test Procedure
· Storage: Room temperature
· Shelf life: 24 months from date of manufacture
Allow test device, urine specimen, and/or controls to equilibrate to room temperature (15-30°C)
prior to testing.
1. Bring the pouch to room temperature before opening it. Remove the test device from the sealed pouch and use it as soon ad possible
2. With arrows pointing toward the urine specimen, immerse the test strip vertically in the urine pecimen for at least 10-15
seconds.Do not pass the maximum line (MAX) on the test strip when mmersing it. See the illustration below
3.Wait for the colored line(s) to appear. The result should be read at 5 minutes. Do not interpret the result after 10 minutes.
NEGATIVE:Two lines appear. One colored line should be in the control region (C),and another colored line should
be in the test region (T).This negative result indicates that cotinine concentration is below the detectable level
NOTE: The shade of the color in the test region (T) may vary,but it should be considered negative whenever
there is even a faint line.
POSITIVE: One colored line appears in the control region (C). No line appears in the test region.This
positive result indicates that the cotinine concentration is above the detectable level (200 ng/mL).
INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques
are the most likely reasons for control line failure. Review the procedure and repeat the test using a new